- Backlit LCD Display
- Stainless Steel Platform For easy clean up
- Locking Mode
- Removable Lid/Tray
- 5 Years Warranty
||g, oz, gn,ct, ozt, dwt
||2 x AAA
||3" x 4.25"
||2.7" x 2.7"
Place your scale on a flat and stable surface away from any electronic equipment that may interfere
with the scale. Press and release the ON/OFF button, wait until the display shows ¡§0.0¡¨ and the zerostable
sign , you may now place the object you want to weigh on the platform. The scale can be
read when the lock sign is shown in the display.
Turn on the scale as described above, place the ¡§tare item¡¨ on the platform, press the ¡§TARE¡¨ button to
zero out the display, when ¡§0.0¡¨ appears on the screen you may now place items on the scale to be
weighed. Once you are finished weighing, remove all objects from the scale, press the ¡§TARE¡¨ again to
cancel the TARE FUNCTION.
RETURN TO ZERO
After weighing an item, if the display does not read ¡§0.0¡¨ press the TARE button to return to ¡§0.0¡¨ and
you are ready to weigh again.
CHANGE WEIGHING MODES
Press the ¡§MODE¡¨ button to change weighing modes on your scale. This scale has 6 weighing modes: g,
oz, gn, ct, ozt, dwt.
With the scale turned on and placed on a stable surface, press and hold the COUNTING & CALIBRATION
button. The display will show ¡§CAL¡¨, press and release COUNTING & CALIBRATION button again and
¡§500.0¡¨ will flash in the display, immediately place a total of 500g weight on the scale. The scale will
reset itself and will give you a PASS message. Calibration is complete.
PARTS COUNTING FUNCTION
Place a small quantity of the item you want to count on the platform, (please note this should be a
quantity of 10, 25 or 5opcs previously counted out by hand.) After the display shows the weight of the
items, then press the ¡§COUNTING&CALIBRATION¡¨ button, the display will automatically show 10, use
the COUNTING&CALIBRATION button to change your quantity selection until it is the same as the
quantity on the scale. The ¡§PARTS¡¨ function will be set up automatically once you have chosen the
quantity and wait for several seconds. Remove the items and the display will show ¡§O¡¨ you may now
count out items as long as their total weight does not exceed 1000g.
Once you are finished with the PARTS COUNTING FUNCTION, press the ¡§MODE¡¨ again to cancel the
The optimum operating temperature for the scale is 50-86¢X F/l 0-30¢X C. Rough handling of the scale will
damage the internal sensor; so you must avoid actions such as shaking or dropping the scale. This is a
sensitive piece of electronic equipment and will be affected by things such as moisture, extreme
temperature changes, magnetic fields, careless handling and misuse.
The scale will automatically